Glove Box Daily Servicing Manual
From Metro Studios Knowledgebase
1) Fill in the use record in time after working the vac glove box box, which include "user, water oxygen benefit, what to do"
two) When replacing gasoline cylinders in the glove box, the pressure is set to the default benefit ± 3mbar
3) The every day functioning pressure is established according to the working comfort and ease. When not in use for a prolonged time, the tension is set to one-6mbar
four) Regularly check out the use of the gloves, test the oil degree of the vacuum pump (concerning the upstream and downstream markings), and add the pump oil correctly
5) Cleanse the water probe at the time every 2 to 3 months to be certain exact detection. (Also need to have to refer to the real scenario of the ecosystem in diverse user bins
six) If you usually use natural solvents in the glove box, it is advisable that you on a regular basis switch the organic solvent adsorbent product, not later on than 1 12 months at the most up-to-date
7) Substitute the filter aspect routinely, no afterwards than two years
eight) According to the precise use of your glove box, decide to swap the vacuum pump oil cycle, this kind of as black and yellow pump oil, h2o or far more natural and organic solvents in the pump oil. The oil alternative cycle ought to not exceed 2 decades at the latest
9) If a huge total of natural solvent is inevitably uncovered through the experiment, the cycle demands to be stopped for the experiment. Soon after the experiment, to start with clear and then start out the cycle
10) When the ecosystem in the box is toxic or dangerous fuel, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and the exhaust port of the cleansing valve should really be exhausted to the outside with an exhaust pipe (when exhaust is not allowed to be outdoors, the exhaust fuel treatment have to be done to start with).